to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||27 December 2018|
|PDF File Size:||6.24 Mb|
|ePub File Size:||6.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Manufactured for and distributed in the U. Quality control recommendation Performing a control test Place the instrument on a level surface or hold it in your hand. This information is stored and must therefore only be mnual once per test strip container. Page 66 Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood. Also always check the battery condition after the display test.
Strip code does not match stored code The test strip belongs to a different test strip lot than the container which was last coded for this parameter. Table of Contents Add to my manuals Add. Press the M button until the desired day is displayed. Open the measurement mqnual flap. Therefore, take the correspond- ing safety measures, such as wearing disposable gloves while working. Alphabetical index Alphabetical index Capillary blood Press the M button to change the currently flashing figure. Page 22 The Accutrend Plus instrument Applying The Control Solution Quality control recommendation Applying the control solution 10 Apply a large free hanging drop of either the glucose or cholesterol low level control solution directly from the bottle to the test strip.
Refer to safety-related notes in the instruc- tions for use accompanying this product.
Page 9 Contents Quality control recommendation Preparing to perform accutredn control test Instrument settings If the displayed setting is correct e. Performing a measurement Blood application in the instrument: Error messages Display Error meaning and description Error: Page 78 Performing a measurement Flagging the measurement as a control: The properties of these test strips are then stored in the instrument.
Repeat the procedure with a new strip. Test strip is unusable The test strip has already been used wccutrend is dirty.
Cobas Accutrend Plus User Manual
Performing A Measurement Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip. Performing a measurement Flagging the measurement as a control: Press the M button to show the next oldest measured value within one memory area.
Cleaning If needed, you can rinse the strip guide separately from the instrument under warm running water. Introduction The Accutrend Plus instrument The Accutrend Plus instrument is used for quantitative measurement of 2 blood parameters: Performing accurtend measurement Now take the test strip from the nanual strip container.
Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed.
Error messages Error messages In certain circumstances error messages manul appear on your display. Be aware the meter is potentially hazardous electronics scrap e-scrap and should be disposed of accordingly.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Memory To navigate within one memory area, use the M button. If the area contains no stored values, two dashes are displayed. The next time the Set button is pressed, the minutes can be set again with the M button. Introduction Test principle By means of a code strip, the instrument reads the lot-specific characteristics of the test strips currently in use.
When you press the button, the place of the measured value within the memory is displayed here: Performing a measurement Code strips The code strip provides the instrument with important information on the specific properties of the respective test strip. Performing a measurement The flashing drop symbol above the strip symbol now instructs you manuql apply the blood. Memory Memory The Accutrend Plus instrument has two memory areas, each of which accutrfnd be used to store up to measured values together with date, ppus and p,us flags.
When the test strip reaches the correct position, you will hear two beeps short — Page 86 Quality control recommendation Press the M button to flag the measurement as a control bottle symbol is displayed.
Regularly check that all display elements are functioning correctly to prevent misinterpretations due to a defective display.
Alphabetical Index Alphabetical index Alphabetical index Avcutrend blood Flap not closed after blood application The measurement chamber flap was not closed properly after blood application; the measurement was not started in time.
Press the M button until the desired month is displayed. Page 55 Performing a measurement The three-digit code number which is also printed on the reverse side of the code strip and on the test strip container is shown in the display. Further information Product specifications Operating conditions and technical data Temperature range for Cholesterol: Flagging Measurements Performing a measurement Flagging measurements You can add additional information to measurements, for example, to characterize a particular value with regard to special circumstances e.
Inserting Batteries Operating the instrument Inserting batteries Performing a measurement Plausibility check after glucose measurement: Note that all handling of blood samples represents a risk of infection.
Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip. B Application area Code strip bottom side Apply the sample here. Performing a measurement Checking the applied blood: If the measured val- ues are to be stored with time information, enter the correct settings see the Setting the date section of this manual.